The back-illuminated FLI camera installed on the spectrograph bench can also be controlled via CCDSH using the regular
sequence commands after loading the necessary module:
Only one camera module can be loaded at a time!
The old water cooler was decommisioned on 11. February 2015. The new OASIS Three chiller (File:Oasis-manual.pdf File:Oasis-productsheet.pdf) is located in the observing room. If necessary, the temperature of the cooler can be queried, set and checked remotely using the ALIX machine using
echo "settemp?" | nc -w 1 172.31.171.98 8821 echo "settemp 10" | nc -w 1 172.31.171.98 8821 echo "temp?" | nc -w 1 172.31.171.98 8821
commands. Fault status can be read by
echo "flts1a?" | nc -w 1 172.31.171.98 8821
A return value of 0 means no error, 2 means "Tank low" warning. For other faults consult the manual. In case of "Pump failure" error the temperature control is stopped, and the cooler must be restarted. The first tests show that at 5 degrees, the cooler tends to stop with "Pump failure", 10C is recommended. Try to keep coolant temperature above the dew point. At this level, the spectrograph can be cooled to -45C with ~95% power level.
If the spectrograph CCD, the cooler, or the ALIX controlling the cooler must be reset, it can be done through the IP Stecker remote-controlled switch, currently only from its website:
The warming-up of the camera should be closely supervised by the observer. Do not turn off the cooling at the operating temperature, warm it up gradually! The regular
ccdsh commands (i.e.
get temperature and
set temperature) can be used.